Dual Trace Oscilloscope, MOS620CH
|High intensity CRT with high acceleration voltage.|
|Trigger level lock function which makes the triggering adjustment unnecessary..|
|Alternate triggering: Even an observation of two waveforms of different frequencies, the waveform of each channel is stably triggered..|
|TV sync triggering: The oscilloscope has a sync separator circuit for triggering of TV-V and TV-H signals.|
|X-Y operation: Set the switch to X-Y. Then the instrument works as an X-Y oscilloscope. CH1 can be applied as horizontal deflection (X-axis) while CH2 provide vertical deflection (Y-axis).|
|Triggering Source||CH1, CH2, LINE, EXT(CH1and CH2 can be selected only when the vertical mode is DUAL or ADD). In ALT mode, if the TRIG.ALT switch is pushed in, it can be used for alternate triggering of two different sources.|
|Coupling||AC: 20Hz to full bandwidth|
|Sensitivity||20Hz~2MHz: 0.5DIV, TRIG-ALT:2 DIV, EXT:200mV
2-20MHz: 1.5 DIV
TRIG-ALT:3 DIV, EXT:800mV
TV: Sync Pulse more than 1 DIV(EXT: 1V)
|Triggering Modes||AUTO: Sweeps run in the free mode when no triggering input signal is applied. (Applicable for repetitive signals of frequency 25Hz or over).
NORM: When no triggering signal is applied, the trace is in the ready state and not displayed.
TV-V: This setting is used when observing the entire vertical picture of television signal.
TV-H: This setting id used when observing the entire horizontal picture of television signal.
|EXT triggering signal input
Input impedance Max. Input Voltage
|Approx: 1M ohm//Approx. 25 pF
400V (DC+AC peak), AC: Frequency not higher than 1KHz.
|Acceleration voltage||Approx. 2kV
|Effective screen size||8×10 DIV (1 DIV=10mm (0.39 in))|
For more details call us on our toll free number.